Welcome to the Bien-Air Dental website
To comply with the new e-Privacy directive, we need to ask for your consent to set the cookies.
REMOVING THE BUR
RINSING THE DEVICE
CLEANING AND DISINFECTION
OPTIONAL: AUTOMATIC DISINFECTION
**For more information, please refer to the manufacturer instructions for use
INTERNAL AND EXTERNAL CLEANING AND DRYING WITH SPRAYNET
Always dry the device.
LUBRICATION WITH LUBRIMED
(For 2-/3-way and 4-way turbines, see inset below)
INTERNAL LUBRICATION WITH LUBRIFLUID FOR 2-/3-WAY AND 4-WAY TURBINES ONLY
Spray into the opening as shown in the illustrations for “2-/3-way” and “4-way”.
PLACE THE TURBINE IN A STERILIZATION BAG
STERILIZE IN ACCORDANCE WITH THE MANUFACTURER‘S INSTRUCTIONS
Only sterilize instruments which have been properly cleaned, dried and lubricated.