Welcome to the Bien-Air Dental website
To comply with the new e-Privacy directive, we need to ask for your consent to set the cookies.
DISCONNECT THE MOTOR FROM THE CABLE
CLEAN UNDER RUNNING WATER AT <38°C
DISINFECTION USING SURFACE DISINFECTANT (follow the manufacturer's instructions for use)
Do not immerse the instrument in a disinfectant bath
PLACE THE MOTOR IN A STERILISATION BAG
STERILISE IN ACCORDANCE WITH THE MANUFACTURER'S INSTRUCTIONS